Three types of neuromonitoring devices:
- Monitors of cerebral hemodynamics (TCD, stump pressure)
- Monitors of cerebral oxygen metabolism (NIRS, SjO2)
- Monitors of cerebral functional state (EEG, evoked potentials)
Awake (nonsedated) patient monitoring is the gold standard of neuromonitoring, but difficult to match during surgery.
Cerebral autoregulation (CAR): maintenance of constant CBF over a range of systemic BP (Figure 99-1)
- Lower limit of autoregulation (LLA): BP under which CBF decreases with BP (here about 50 mm Hg)
- Upper limit of autoregulation (ULA): BP above which CBF increases with BP (here about 150 mm Hg)
- CAR shift: modification of LLA and/or ULA set points, associated with altered relationship between systemic BP and CBF (LLA right shift in chronic HTN and anemia)
Figure 99-1. Autoregulatory Curve
Flow is generally shown as being stable between mean arterial pressures (or more strictly, perfusion pressure) of 50 and 150 mm Hg. However, although this expresses the concept of autoregulation well, it does not depict reality. Reproduced from Longnecker DE, Brown DL, Newman MF, Zapol WM: Anesthesiology. Figure 50-10A. Available at: www.accessanesthesiology.com. Copyright © The McGraw-Hill Companies, Inc. All rights reserved.
- Measures cerebral blood velocity (Vx), which correlates with cerebral blood flow (CBF)
- Relative changes over time are more accurate than absolute values
- Three main sites to obtain Doppler signal (Figures 99-2 and 99-3): temporal (most common, used essentially to measure Vx in MCA), suboccipital (posterior cerebral circulation), orbital (anterior cerebral circulation)
Figure 99-2. Sites to Obtain TCD Signal
Reproduced from Levitov A, Mayo PH, Slonim AD: Critical Care Ultrasonography. Figure 26-19. Available at: www.accessanesthesiology.com. Copyright © The McGraw-Hill Companies, Inc. All rights reserved.
Figure 99-3. Circle of Willis
Depths of insonation from ipsilateral transtemporal window. Reproduced from Levitov A, Mayo PH, Slonim AD: Critical Care Ultrasonography. Figure 26-20. Available at: www.accessanesthesiology.com. Copyright © The McGraw-Hill Companies, Inc. All rights reserved.
Problems encountered with TCD monitoring: no acoustic window (5%–15%), probe dislocation
- Peak systolic velocity (PsVx) and End-diastolic Velocity (EdVx): measured directly
- Mean velocity (MVx): derived from PsVx and EdVx
- Pulsatility index (PI): maximal variation of Vx from systole to diastole, weighted by MVx
- Used to study changes in distal cerebral vascular resistance if HR and systemic BP pulsatility are maintained constant
Factors Affecting Velocity
|Factors associated with increased Vx signals||Factors associated with decreased Vx signals|
|Anesthetic inhalational agents||Anesthetic induction agents (except ketamine)|
|Vasopressors, hypertension with loss of CAR||Vasodilators, hypotension with loss of ...|
Log In to View More
If you don't have a subscription, please view our individual subscription options below to find out how you can gain access to this content.
Want remote access to your institution's subscription?
Sign in to your MyAccess profile while you are actively authenticated on this site via your institution (you will be able to verify this by looking at the top right corner of the screen - if you see your institution's name, you are authenticated). Once logged in to your MyAccess profile, you will be able to access your institution's subscription for 90 days from any location. You must be logged in while authenticated at least once every 90 days to maintain this remote access.
If your institution subscribes to this resource, and you don't have a MyAccess profile, please contact your library's reference desk for information on how to gain access to this resource from off-campus.
AccessAnesthesiology Full Site: One-Year Subscription
Connect to the full suite of AccessAnesthesiology content and resources including procedural videos, interactive self-assessment, real-life cases, 20+ textbooks, and more
Pay Per View: Timed Access to all of AccessAnesthesiology
24 Hour Subscription $34.95
48 Hour Subscription $54.95
Pop-up div Successfully Displayed
This div only appears when the trigger link is hovered over.
Otherwise it is hidden from view.