Comparison of Available Noninvasive Cardiac Output Monitors
|CO2 Fick principle, for example, NICO®|
- Good agreement between NICO® and ultrasound transit time
- Patient may be breathing spontaneously or on a ventilator
- Accuracy adversely affected by hyperventilation and V/Q mismatch (preexisting lung disease or postoperative atelectasis)
|Yes; however, patient must be intubated|
|Esophageal Doppler, for example, CardioQ (single use), WAKIe TO (reusable)|
- High validity for monitoring changes in cardiac output in critically ill patients
- Using esophageal Doppler to optimize fluid management has been shown to improve outcomes
- Patient must be intubated and sedated
- Small risk of injury to the esophagus
- Limited agreement with thermodilution
|No, requires placement of probe in the esophagus|
|Transpulmonary lithium dilution, for example, LiDCO|
- High signal-to-noise ratio
- Good agreement with thermodilution
- Low risk of toxicity
- Not approved for patients <40 kg, pregnant women, or patients taking lithium
- Accuracy reduced by aortic insufficiency, intra-aortic balloon pump, nondepolarizing NMB
|No, requires an intra-arterial line|
|Arterial pulse contour analysis, for example, Vigileo®|
- Good agreement with other methods of cardiac output determination
- Continuous measurement of SV and CO
- Also display SVV (SV variation) with a value >15% indicating fluid responsiveness
- Easy to use
- Adversely affected by a poor arterial waveform (arrhythmia, intra-aortic balloon pump)
- Older devices may not adequately compensate for changes in vascular tone
- Newer devices have good agreement with thermodilution and esophageal Doppler
|No, requires an intra-arterial line|
|Bioimpedance, for example, Cheetah NICOM®, BioZ Dx®|
- Completely noninvasive
- Impedance to high-frequency current between electrodes on chest; estimates thoracic blood volume
- Relative hemodynamic stability required
- Accuracy reduced by arrhythmias, aortic insufficiency, intra-aortic balloon pump
- Contraindicated in patients with pacemakers
|Yes, adhesive surface electrodes only|
- The indirect Fick equation is defined as the following:
where V̇CO2 is the CO2 produced, CvCO2 is the central venous CO2 content, and CaCO2 is the arterial CO2 content. V̇CO2 is determined by the minute ventilation and instantaneous CO2 content. CaCO2 is estimated from the end-tidal CO2.
- By combining measurements during rebreathing and during normal ventilation, CvCO2 can be eliminated from the equation
- To improve accuracy, shunt fraction can be estimated from FiO2 and Pao2 from blood gases
- A Doppler transducer is inserted into the esophagus of an anesthetized intubated patient such that the characteristic waveform of blood flow in the descending aorta is displayed
- The area under the aortic velocity envelope is calculated as a velocity time integral (VTI) that is referred to as the stroke distance (SD)
- The monitor utilizes a computer algorithm to determine the stroke volume (SV) and CO by multiplying the SD by the cross-sectional area of the descending aorta (Figure 41-1)
Figure 41-1. Typical Esophageal Doppler Descending Aortic Waveform
Log In to View More
If you don't have a subscription, please view our individual subscription options below to find out how you can gain access to this content.
Want remote access to your institution's subscription?
Sign in to your MyAccess profile while you are actively authenticated on this site via your institution (you will be able to verify this by looking at the top right corner of the screen - if you see your institution's name, you are authenticated). Once logged in to your MyAccess profile, you will be able to access your institution's subscription for 90 days from any location. You must be logged in while authenticated at least once every 90 days to maintain this remote access.
If your institution subscribes to this resource, and you don't have a MyAccess profile, please contact your library's reference desk for information on how to gain access to this resource from off-campus.
AccessAnesthesiology Full Site: One-Year Subscription
Connect to the full suite of AccessAnesthesiology content and resources including procedural videos, interactive self-assessment, real-life cases, 20+ textbooks, and more
Pay Per View: Timed Access to all of AccessAnesthesiology
24 Hour Subscription $34.95
48 Hour Subscription $54.95
Pop-up div Successfully Displayed
This div only appears when the trigger link is hovered over.
Otherwise it is hidden from view.