The most sensitive leads for ST segment monitoring are V3, V4, and V5.
Sensitivity of Various Lead Combinations for Detecting Ischemia
|London et al.||Landesberg et al.|
|II + V2 + V3 + V4 + V5||100%|
|V4 + V5||90%||V3 + V5||97%|
|II + V5||80%||V4 + V5||92%|
|II + V4 + V5||96%||V3 + V4||100%|
|Single leads V5, V4, and V3||75%, 61%, and 24%, respectively||Single leads V5, V4, and V3||75%, 83%, and 75%, respectively|
- Cardiac events are the most common cause of death in the perioperative period
- Unstable angina, myocardial infarction, and heart failure are estimated to occur in 1–10% depending on the risk of the surgical procedure and the severity of patient’s disease
- Overall, 4% of patients undergoing noncardiac surgery have significant cardiac events
May occur in 84% of patients during surgery.
Etiology: ischemia, anesthetics, electrolyte abnormalities, endotracheal intubation, reflexes, vagal stimulation, central nervous stimulation, intracranial hemorrhage, autonomic nervous system dysfunction, central venous cannulation, surgical manipulation of cardiac structures, pericardial effusion, hypothermia, acute pericarditis, pulmonary embolus.
- Electrodes are placed on the right arm (RA), left arm (LA), and left leg (LL)
- For bipolar leads (I, II, and III), one pair is selected for monitoring and the other one is used as a ground (i.e., the electrodes are positive, negative, and ground)
- While the three-lead system is useful for arrhythmia detection, it is of limited utility when monitoring for myocardial ischemia
- The modified three-electrode system includes modified chest leads (MCL), CS5, CM5, CB5, and CC5
- P waves are maximized for atrial dysrhythmia monitoring and there is increased sensitivity for detecting anterior wall ischemia
Figure 39-1. Modified Three-Lead Systems to Detect Ischemia
MCL1: monitor lead III with LL electrode placed on right fifth intercostal space, on the medioclavicular line; CS5, monitor lead I with LA electrode placed on left fifth intercostal space on the anterior axillary line; monitor lead II with LL electrode in its normal position (for arrhythmia); CM5, monitor lead I with RA electrode on sternal manubrium and LA electrode on left fifth intercostal space on the anterior axillary line; CB5, monitor lead I with RA electrode over right scapula and LA electrode on left fifth intercostal space on the anterior axillary line; CC5, monitor lead I with RA electrode on right fifth intercostal space on the anterior axillary line and LA electrode on left fifth ...
Please Try Again Later
Your institution has exceeded the maximum number of simultaneous users. Please try later.
Want remote access to your institution's subscription?
Sign in to your MyAccess profile while you are actively authenticated on this site via your institution (you will be able to verify this by looking at the top right corner of the screen - if you see your institution's name, you are authenticated). Once logged in to your MyAccess profile, you will be able to access your institution's subscription for 90 days from any location. You must be logged in while authenticated at least once every 90 days to maintain this remote access.
Create a Free MyAccess Profile
Note: If you have registered for a MyAccess profile on any of the Access sites, you can use the same MyAccess login credentials across all sites.
Benefits of a MyAccess Profile:
- Remote access to the site off-campus on any device
- Notification of new content via custom alerts
- Bookmark your favorite content such as chapters, figures, tables, videos, cases and more
- Save and download images to PowerPoint
- Self-Assessment quizzes saved for quick review
- Custom Curriculum access for both instructors and learners
AccessAnesthesiology Full Site: One-Year Subscription
Connect to the full suite of AccessAnesthesiology content and resources including procedural videos, interactive self-assessment, real-life cases, 20+ textbooks, and more
Pay Per View: Timed Access to all of AccessAnesthesiology
24 Hour Subscription $34.95
48 Hour Subscription $54.95
Pop-up div Successfully Displayed
This div only appears when the trigger link is hovered over.
Otherwise it is hidden from view.