Normal physiologic hemostasis is achieved through the interplay of pro- and anti-coagulant properties of the vascular endothelium, platelets, and the coagulation cascade. Conventional laboratory tests look at this system in a piecemeal fashion and give a limited indication of in vivo functionality. These tests take significant time for results, further limiting real-time usefulness.
To address these limitations, anesthesiologists have adopted the use of thromboelastography (TEG). TEG provides a rapid point-of-care assessment of the coagulation process, with initial results available within 5 minutes. TEG measures viscoelastic changes of clot formation through clot lysis, evaluating the integrity of the coagulation system, platelet function, fibrin polymerization, clot strength, and fibrinolysis. Results of TEG permit targeted transfusion. TEG is used in liver transplant, cardiac bypass, and, increasingly, trauma and general surgery.
TECHNIQUE AND INTERPRETATION
The TEG analyzer consists of an electronic recorder attached to a torsion wire that suspends a plastic pin into a plastic cup or cuvette. A 0.36-mL whole blood sample is mixed with an activator, such as kaolin or celite, to trigger clot formation. The sample is pipetted into the cup, and the cup begins to oscillate slowly through an angle of 4° at a temperature of 37°C. Initially, the movement of the cup does not affect the pin, but as the blood thickens, the pin becomes entangled in the developing clot, coupling the motion of the cup to the pin. The movement of the pin is transferred through the torsion wire to the electronic recorder, and the analyzer produces a tracing (Figure 7-1).
Thromboelastogram tracing. (Reproduced, with permission, from Kashuk JL, Moore EE, Sawyer M, et al. Postinjury coagulopathy management: goal directed resuscitation via POC thrombelastography. Ann Surg. 2010;251:604.)
Several parameters are presented on the tracing:
Reaction time—The time in minutes elapsed from the start of the test until the clot moves the pin enough to produce a 2-mm amplitude on the tracing is defined as the reaction time (R). R reflects the activity of the coagulation cascade; a coagulation factor deficiency produces a prolonged R and hypercoagulability yields a shortened R time. The normal values for R depend on the type of clotting activator used.
Alpha angle—The alpha angle (α) is a measure in degrees of the speed of clot formation. It is defined as the angle between the horizontal axis of the tracing and the tangent to the tracing at 20-mm amplitude. Decreased angles indicate a slower rate of clot strengthening, as seen with low fibrinogen levels. Normal values are in the range of 45°–55°.
Coagulation time—The coagulation time (K) is measured in minutes from the end of R to when the tracing amplitude reaches 20 mm. Like the alpha angle, K is determined by the rate at which the ...
Log In to View More
If you don't have a subscription, please view our individual subscription options below to find out how you can gain access to this content.
Want remote access to your institution's subscription?
Sign in to your MyAccess profile while you are actively authenticated on this site via your institution (you will be able to verify this by looking at the top right corner of the screen - if you see your institution's name, you are authenticated). Once logged in to your MyAccess profile, you will be able to access your institution's subscription for 90 days from any location. You must be logged in while authenticated at least once every 90 days to maintain this remote access.
If your institution subscribes to this resource, and you don't have a MyAccess profile, please contact your library's reference desk for information on how to gain access to this resource from off-campus.
AccessAnesthesiology Full Site: One-Year Subscription
Connect to the full suite of AccessAnesthesiology content and resources including procedural videos, interactive self-assessment, real-life cases, 20+ textbooks, and more
Pay Per View: Timed Access to all of AccessAnesthesiology
24 Hour Subscription $34.95
48 Hour Subscription $54.95
Pop-up div Successfully Displayed
This div only appears when the trigger link is hovered over.
Otherwise it is hidden from view.