Chapter 9

Introduction

Objectives

1. Apply the concept of time constant to the physiology of mechanical ventilation.

2. Compare constant flow and descending ramp flow patterns during volume-controlled ventilation.

3. Describe the effect of respiratory mechanics on the airway pressure waveform during volume-controlled ventilation.

4. Describe the effect of resistance and compliance on flow during pressure-controlled ventilation.

5. Describe the effect of rise time adjustment during pressure-controlled and pressure support ventilation.

6. Describe the effect of termination flow during pressure support ventilation.

7. Discuss the role of sigh breaths during mechanical ventilation.

8. Discuss the physiologic effects of I:E ratio manipulations.

Microprocessor-controlled ventilators allow the clinician to choose among various inspiratory flow waveforms. This chapter describes the technical and physiologic aspects of various inspiratory waveforms during mechanical ventilation.

Time Constant

An important principle for understanding pulmonary mechanics during mechanical ventilation is that of the time constant. The time constant determines the rate of change in the volume of a lung unit that is passively inflated or deflated. It is expressed by the relationship:

where Vt is the volume of a lung unit at time t, Vi is the initial volume of the lung unit, e is the base of the natural logarithm, and τ is the time constant. The relationship between Vt and τ is illustrated in Figure 9-1. Note that the volume change is nearly complete in five time constants.

Figure 9-1

The time constant function for lung emptying. After one time constant, 37% of the volume remains in the lungs, 13% remains after two time constants, 5% remains after three time constants, 2% remains after four time constants, and < 1% remains after five time constants.

For respiratory physiology, τ is the product of resistance and compliance. Lung units with a higher resistance and/or a higher compliance will have a longer time constant and require more time to fill and to empty. Conversely, lung units with a lower resistance and/or compliance will have a shorter time constant and thus require less time to fill and to empty. A simple method to measure the expiratory time constant is to divide the expired tidal volume by the peak expiratory flow during passive positive pressure ventilation:

where VT is the expired tidal volume and V̇e(peak) is the peak expiratory flow. Although this is a useful index of the global expiratory time constant, it treats the lung as a single compartment and thus does not account for time constant heterogeneity in the lungs.

Flow Waveforms

Volume-Controlled Ventilation

The flow, pressure, and volume waveforms produced with a constant flow pattern are shown in Figure 9-2. This is often called square-wave or rectangular-wave ventilation due to the shape of the flow waveform. With the constant flow pattern, the volume ...

Sign in to your MyAccess profile while you are actively authenticated on this site via your institution (you will be able to verify this by looking at the top right corner of the screen - if you see your institution's name, you are authenticated). Once logged in to your MyAccess profile, you will be able to access your institution's subscription for 90 days from any location. You must be logged in while authenticated at least once every 90 days to maintain this remote access.

Ok

Subscription Options

AccessAnesthesiology Full Site: One-Year Subscription

Connect to the full suite of AccessAnesthesiology content and resources including procedural videos, interactive self-assessment, real-life cases, 20+ textbooks, and more